Important Announcement: Community Back to Full Functionality
Dear Members, we are thrilled to announce that our Community is back to full functionality and that posts publication is now enabled again! We appreciate your patience during the last weeks. Learn more about our Community Guidelines. Thank you, Schneider Electric Community Team.
Ask: How step by step to do accuration verification on ION8650 meter using Reference Meter
Metering & Power Quality
Schneider Electric support forum about Power Meters (ION, PowerTag, PowerLogic) and Power Quality from design, implementation to troubleshooting and more.
Link copied. Please paste this link to share this article on your social media post.
Thanks for your respond and also the link for reference.
I havetried,butstillcouldnotdue tothe position of"wattinfraredpulser"is moreprominentthan the"LEDpulseindicator", while the Zera meter probesurface is "flat". Or, may be this is note the problem. Need more advice.